When Auto-lid is on, the lid of the centrifuge will open automatically at the end of a run.
This will tell you how long your sample stays in the chamber after a run has been completed.
The Short/Spin button starts the centrifuge running and it will continue to run as long as the button is held down. A short spin is usually used to collect the liquid in the tube. The short spin speed can be set in the System.
The Short/Spin speed is set via the System. Press “Speed” on the main screen. Turn the knob to the required speed and press enter. Then find the “SPIN” function in the “System” and select “On” to confirm.
This is a safety feature to ensure that the lid has been closed intentionally.
Yes, it is normal. There is a cooling fan in the lid and the chamber is sealed when the lid is locked.
Yes, it is normal. The fan in the centrifuge cools the chamber to prevent heat accumulation. The fan continues to run at a low speed when the rotor is not running.
These slits allow excessive heat to escape when the centrifuge is running at a high speed and this keeps the heat in the chamber to a minimum.
Yes, you can change the speed and time during centrifugation.
Yes. You can set the centrifuge to 9 different ramps. The acceleration and deceleration times can be set from 15 to 90 seconds.