Standard 10x objective lens included; 20x objective lens also available for optional purchase. Users can switch lenses based on their applications.
The FOV range of the 10X objective lens is 2.6 mm X 2.0 mm, and the FOV range for the 20X objective lens is 1.3 mm X 1.0 mm.
EzScopes has been rigorously tested and can endure such environment. However, in order to reduce additional corrosion caused to the device due to condensed water droplets, we strongly recommend users leave EzScope for 1-2 hours after placing it in the incubator and wait for the temperatures of the machine and incubator to balance, and then connect the power of the machine and place samples inside for observation. Doing so can also reduce the temperature changes when the cell sample comes in contact with the carrier of the machine.
There will be very little impact. Of course, the most perfect cable guiding method is through the dedicated equipment port for the incubator. But if your incubator does not have an equipment port, you can still clamp the signal cable and power cable between the door and the hermetic gasket. The hermetic gasket in the accessory pack of EzScope can be used when necessary to coat where the cable is clamped in order to ensure there is no leakage at all, and further protect the cable.
You can simply wipe with the regular 70% alcohol in laboratories. Please beware not to use too much disinfectant and prevent liquids from leaking inside the machine.
You can first blow away the dust on the lens surface with a blower, and then drop 1-2 drops of 70% alcohol on the lens tissue and wipe the lens lightly. Wipe from the center of the lens outward in a spiral path, and then use a new lens tissue to wipe off the residual liquid.
Yes; EzCapture is equipped with measuring functions; please refer to the operation manual for details.
We’ve always insisted on continual improvement and providing more better products. We will strive to provide more analysis functions; you can update the software for free from the official website of Blue-Ray Biotech in the future. In addition, there are many 3rd party software systems available on the market that can also be used. For example:Fiji (https://imagej.net/Fiji) or CellProfiler (https://cellprofiler.org/).
EzCapture only supports the Win 10 system. Please confirm the version of your computer system, install and execute the software on the correct system.
EzCapture will save the captured pictures on the control computer connected to EzScope. There are also many third-party remote access software systems available for use (please refer to the operation manual for details); you can use these software systems to access the pictures on the computer remotely. Please note that operations can only be performed under environments with internet connections.
When the conditions of the incubator components are poor or when there is a vibration source nearby, it may cause some or all of the images in the time-lapse series video to become blurry; please first eliminate the vibration source.
Small Tip: In time-lapse recorded videos, you can first use the built-in video editing functions to remove the blurry images before outputting the video.